Cell viability assays are utilized in a wide variety of research areas ranging from examining the mechanisms of cell death and cell proliferation to evaluating the effects of cytotoxic compounds and novel therapies, including drug candidates, pathway activators and inhibitors, and genetic reporters. High-quality image acquisition combined with complete multiparameter, high-throughput screening provides an optimal method to assess cell viability. In this application note, we report the use of the EarlyTox™ Live/Dead Assay Kit in conjunction with the ImageXpress® Micro High-Content Imaging System and the MetaXpress® High-Content Image Acquisition and Analysis Software.
The EarlyTox Live/Dead assay kit contains a marker for live cells and a marker for dead cells, both of which are suitable for use with mammalian cells. Their unique mechanisms of action are dependent upon cell membrane permeability. Viable cells are stained with intense green fluorescence in the cytosol, with Calcein AM, a widely used live cell marker. The non-fluorescent Calcein AM permeates the intact cell membrane where the acetoxymethyl (AM) group is cleaved by intracellular esterases, thus yielding the fluorescent calcein molecule. Calcein AM can be used as a standalone reagent for cell proliferation assays or other assays where only live cell staining is desired, as provided in the EarlyTox™ Live Cell Assay Kit (Molecular Devices P/N R8342 for Explorer kit, P/N R8343 for Bulk kit).
Ethidium homodimer-III (EthD-III) is virtually non-fluorescent and impermeant to an intact plasma membrane. In the event of compromised membrane integrity that is associated with cell death, EthD-III enters dead cells and binds to nucleic acids, resulting in bright red fluorescence. Cytotoxic events that affect cell membrane integrity and that eventually lead to cell death can be accurately assessed using this method.
The assay kit enables characterization of a full concentration-response profile of test compounds since the no-wash, homogeneous assay protocol eliminates the potential to remove dead and dying cells. Fluorescent signals from calcein and EthD-III can be detected using the ImageXpress Micro system and rapidly analyzed using the standard Live/Dead application module in MetaXpress software.
- EarlyTox Live/Dead Assay Kit
- Explorer Kit (2-plate size, Molecular Devices P/N R8340)
- Bulk Kit (10-plate size, Molecular Devices P/N R8341)
- HeLa cells (ATCC P/N CCL-2)
- HeLa media
- Minimum Essential Medium complete media with glutamine and serum
- CHO-M1WT3 cells (ATCC P/N CRL-1985)
- CHO media
- Ham’s F-12 complete media with glutamine and serum
- Staurosporine (Sigma P/N S5921)
- Mitomycin C (Sigma P/N M4287)
- 96-well black, clear-bottom microplates (Greiner P/N 655090)
- 384-well black, clear-bottom microplates (Corning Falcon P/N 62406-490)
- ImageXpress Micro High-Content Imaging System with MetaXpress High-Content Image Acquisition and Analysis Software
HeLa cells were plated in a 96-well microplate at 6,000 cells per well in a volume of 100 µL. In a 384-well microplate, CHO-M1 cells were plated at 5,000 cells per well in a volume of 50 µL. Both plates were placed in an incubator at 37°C, 5% CO2 overnight to allow for cellular adhesion and growth. Both cell types were then treated for 24 hours with staurosporine (general protein kinase inhibitor and a potential anti-cancer therapeutic).
HeLa cells were also treated with mitomycin C (potent DNA crosslinker and chemotherapeutic). The HeLa cells were treated in quadruplicate with compounds serially diluted 1:3, while the CHO cells were treated in triplicate with staurosporine serially diluted 1:2. 100 µL of 2X compound dilutions were added to corresponding wells in the HeLa plate, except for control wells. The final concentrations of staurosporine were 1 µM down to 0.000017 µM (0.017 nM), and the final concentrations of mitomycin C were 100 µM down to 1.694 µM. 50 µL of 2X staurosporine dilutions were added to the wells in the CHO plate for final concentrations of 2 µM down to 0.00024 µM (0.24 nM). Serially diluting the compounds enabled analysis of dosedependent responses to the cytotoxic effects of these compounds.
A 2X stain working solution was prepared by combining Calcein AM and EthDIII stock solutions in warm DPBS for a concentration of 2 µM for each dye. Prior to the addition of the stain solution, an equal volume of media was removed from each well. This resulted in final concentrations of 1 µM for each dye. The plates were incubated at 37°C, 5% CO2 for 45 minutes. In addition to the EarlyTox Live/Dead kit, a nuclear stain such as Hoechst 33342 can be added to the cells to aid in acquiring total cell count data. After the incubation for the Live/Dead stains, 10 µL of a 55 µM (11X) Hoechst working solution was spiked into each well for a final concentration of 5 µM per well. The cells were incubated at 37°C, 5% CO2 for 15 minutes longer.
Immediately after the final incubation, the plates were imaged on the ImageXpress Micro system using a 10X Plan Apo objective and the FITC, Texas Red, and DAPI (since the nuclei were counterstained with Hoechst) filter sets. At this magnification, one field-of-view is capable of capturing 1,100-1,500 cells so that a single image yields statistically relevant results. The HeLa cell plate was imaged at two sites per well, which enabled analysis of >2000 cells/well. Images were taken using the widefield imaging mode (Figures 1 & 2).
Bright staining enables fast read times
Capturing 10X images using the ImageXpress Micro system was highly efficient, as it took 2 ½ minutes to run an entire 96-well plate and 9 ¼ minutes to run a 384-well plate in 2 colors (FITC, Texas Red). The bright intensity of the fluorophores in the kit allowed the use of short exposure times, which enabled rapid image acquisition and prevented unwanted photobleaching.
Preconfigured image analysis modules facilitate accurate data generation
The widefield images were analyzed using the Live/Dead application module in MetaXpress software. With intuitive user input, the module was configured to identify and differentiate live and dead cells based on the stain each cell type expressed. Multiple measurement parameters, such as the number of live and dead cells, percent of live and dead cells, total cell count, and intensity of each wavelength could be assessed. Furthermore, if nuclei are counterstained with Hoechst, a Multi-Wavelength Cell Scoring software module can be run along with (or instead of) the Live/Dead module. This provides additional analyses for the cell viability assay, such as scoring cells positively fluorescing at specific wavelengths (FITC or Texas Red) as well as accurately counting all nuclei.
Toxicity EC50 values can be calculated from dose-response curves
Live and dead cells were imaged and quantitative analysis was run based on cells staining positive for either calcein (green fluorescence) or EthD-III (red fluorescence) (Figures 1 & 2). Treatments of HeLa cells with staurosporine and mitomycin C both showed clear concentration responses (Figure 3). Graphs for percent of live cells and percent of dead cells vs. compound concentration produced dose response curves with EC50 values of 0.121 µM for staurosporine and 9.307 µM for mitomycin C. The same can be said for the CHO cells treated with staurosporine, as they too exhibited a distinct dose-dependent response (Figure 4). The curves for the percent of live and dead CHO cells vs. staurosporine concentration generated EC50 values of 0.180 µM. The comparable EC50 values for the staurosporine treatments in both experiments further demonstrates the accuracy of the EarlyTox reagents in the determination of live and dead cells.
The EarlyTox Live/Dead Assay Kit, analyzed with the ImageXpress Micro system with MetaXpress software, enables an accurate measurement of live and dead cells with a highly efficient workflow. The brightness of the fluorophores allow the scientist to optimize for low reagent concentrations while still enabling fast and robust image-based quantitation of the cell viability markers from the EarlyTox Live/ Dead kit. The consistent and statistically relevant quantitative analysis allows for the testing of multiple compounds at differing concentrations and can be adapted to nearly any cell viability assay.